This is an old revision of the document!
To switch to this instrument click on the Logic Analyzer button from the left menu.
This instrument can run continuously, repeatedly triggering, by pressing the Run button or triggering once, by pressing the Single button.
The logic analyzer instrument consists of a channel manager, a central signal plot and a control panel for different settings.
The channel manager allows the user to enable or disable channels, set triggers, create groups of signals or set protocol decoders to groups. The groups can be created by selecting channels and pressing the button or by dragging one channel/group over another channel/group. The drag and drop mechanism also allows the user to reorder the channels.
Channel manager configuration for a pin:
Channel manager configuration for a pin belonging to a group:
Channel manager configuration for a group:
The plot displays the signal data captured on the enabled channels and the decoded data for active groups.
The control panel can be opened by pressing the button of any element from the right side of the bottom menu bar, by using the General Settings button or the Channel Settings button . The available elements in the bottom menu bar are Trigger Settings and Cursor Settings.
If a channel is highlighted in the channel manager, the settings for that particular channel are displayed in this panel. All the channels/groups allow the user to change the name by writing a value in the edit box.
Return to Scopy Main Page