To prepare the SD-card for the CoraZ7-07s board:
Once microSD card has been imaged, safely remove the hardware from the SD card writer, and insert the card directly into the microSD card slot on the CoraZ7-07s.
Down the latest IIO-Oscilloscope release from Github, and install it on your PC. (You may need to right-click the installer, and run as “Elevated” in order to get it to install. )
|Selects how the CoraZ7-07s board boots. We want to boot from the microSD card
|Across Pins 1 & 2
|Selects how the CoraZ7-07s board is powered. We are powering from the USB port
|Across Pins 3 & 2 (labeled USB)
Connect the power, cables, and sensor according to the diagram below.
Once the microSD card and cables have been connected to the CoraZ7-07s and CN0540 its now time to boot the system.
If you don't have a network available and want to stream data directly from the Ethernet port of the CoraZ7-07s to the Ethernet port of your PC that is still possible, but requires some extra configuration. Please see the Network Configuration page for complete details.
Before you can start streaming data, you first must locate the CN0540 on your network.
Now its time to start communicating with the CN0540 so you can start streaming data. When you first open IIO-Oscilloscope you'll see two windows.
The CN0540 IIO Plugin automatically configures the CN0540, so it is ready to use as soon as you run the application. Calibration of the sensor is also automatically performed so that a user can start using the capture window to collect and analyze data. No other configuration is required for the application.
If you want to re-calibrate the system, shut the system down, or modify individual registers of the devices on the CN0540 that can also be done either using the CN0540 Plugin or using the DEBUG panel to write/read specific registers. This is optional and typically application specific.
Below is a picture of what the CN0540 IIO Plugin looks like.
|Checks the current status of the ADG5421's FF Pin
|Shutdowns power from the AD7768-1
| Disabled(Power On)
|ADC Driver Settings
|ADA4945 Operational Status
|ADA4945 Power Mode
| Checked(Full Power)
|The Calibration is set for a 10V calibration
|Calibrated AD7768-1 Input voltage offset
|Calibrated LTC2606 Level Shifting Voltage
|Calibrated Sensor Input voltage
|Input voltage coming from the sensor
|ADA4945 Common Mode input voltage
|AD7768-1 Ain+ input voltage
|AD7768-1 Ain- input voltage
For CbM applications, most customers are typically interested in the frequency domain plots. To obtain a FFT plot, do the following:
End of Document